You Save: Rs. 3,400.00 (65%)
Generally Medical air mattresses used with Medical Air Bed also known as Air Hospital Beds. These medical air mattresses also searched for Chirayu air bed that play a pivotal role in preventing and treating, Serious ailments related to extended continous bed ridden or bed rest, such as pressure sores and skin shearing.
Heavy duty Air mattress, basic treatment therapeutic mattress supports and good for pressure ulcers therapy, alternative pressure anti bedsore air bed mattress.
Chirayu this original Sanskrit word conveys the meaning that the universal human desire for longevity.
It is also blessing that is showered by elders when their loved ones come to pay their respects Alternating pressure mattress therapy is an ideal solution for bedsores prevention and care.
Advance technology, heavy duty motor, quiet, and adjustable pump allows individuals for comfort setting.
A heat sealing by uniform tooling bubble mattress ensure comfortable performance both for the patient as well as the care taker and makes.
Reliable and easy to Use, Aluminum motor cover, Low noise, Low Vibration, Low temperature, Easy Adjustable pressure with knob Hook for easy fit on bed frame
Replacement Warranty On Pump: 1 year
Air Bubble Mattress Specifications
Max User Weight 130-150 Kg
Dimensions: 200 x 90 x 7 cm
Cells Material: PVC En71.3mm
Compact Adjustable Pump specifications
Power supply: 220 V – 240
V Cycle time: 6 mins
Air Output: 6-7 L/min
Pressure Range: 35 – 130 mmHg
Protection Fuse 500 mA
Place the pump of air mattress on a flat table or hand it with the help of the hanger at the foot side. Unfold & Lay the bubble air mattress on a foam mattress in a way that the connections are on the foot side. The excess ends on the leg side can be tucked under the foam mattress. On the head side excess air mattress can be used to make a pillow or can be tucked in
Step 2 Connect one end of the air tubes with the air pump and the other end with the air bubble mattress. Connect them firmly at both the ends make sure that air pipes are no clinked.
Switch on the electric pump on the air mattress by connecting with 220-V AC. The air mattress will start inflating. Keep the pressure adjusting knob at full in the beginning. The pump should be constantly switched on during the usage otherwise the mattress will deflate. A type bubbles will inflate rather B type will remain deflated starting the process. Adjust pressure ideally with the adjusting knob in such a way that there is a two fingers gap between the deflated air bubbles and the patient.
Step 4 To repack the mattress when not in use, you need to exhale the air out to the nozzles by pressing it from other side.
In case the pump of air mattress does not get switched on please check the fuse at the bottom of the pump of air mattress, if burn replace with a new fuse of 1.5 A/240V. If the pump is working but the mattress is not inflating please check the connections and then check the air mattress for leakage. If found punctured use the puncture kit provided with the air mattress. Note • Keep sharp objects away from the air mattress
• Do not stand on the air mattress
• Keep the pump away from water, damp surface.